These ventilators have had several high profile uses, from Christopher Reeve to injured servicemen in Iraq.
Composite cylinders provide several benefits in this application. Dimensional stability was the core factor in replacing the original aluminum cylinder. Polyslide's lower weight is also desirable in portable devices.
In such a critical application, seal life is a important concern. Metal cylinders would require a high RA finish to insure longevity, further increasing cost. PolySlide's smooth, rounded surface is seal-friendly, promoting long life and reduced operational friction. A cup seal is employed to further reduce friction and improve operational life-cycles.
Combating Static Build-up
Composites often pose new and unconventional engineering challenges with their unique material properties. Initial prototypes would fail after only a few cycles due to static build-up. The aspirator assembly became in essence a Van de Graaff static generator due to the dielectric nature of composite tubing.
Combined with the dangers of fire when used with oxygen supplementation, this was a major project hurdle. Polygon's engineering team produced a hallmark solution by including conductive fibers to safely dissipate the static build-up.
Polygon has also integrated improvements from this project into the standard product line, such as eliminating the need for gelcoat, improving cylinder ID finish to further reduce friction, and more.
Customization and Weight Reduction
Polygon has also updated the assembly provided through the years to add other improvements such as embedded metal screw mounting points for easier replacement and routine maintenance, and converting the piston and rod to composite materials for greater weight savings. These weight savings produced two benefits to the product. Lower weight means less inertia to overcome when switching piston direction, and thus lower operational force. This allowed further weight savings via miniaturization of power linkages.
The other benefit is a reduction in operational noise. Each change in piston direction produces a clicking sound from pressure change and valve action. More than just being intrusive, this repeating sound can become maddening similar to a clock in an empty room. Composites are lightweight and less resonant, producing a more pleasant experience for users of this full-time lifeline.
Any ratings are typical for design purposes. Your design parameters may affect final ratings. Consult with a Polygon sales engineer for guidance. Final testing and approval is the customer's responsibility for their application. This information is derived from our testing and published data. There is no assurance of these properties, or warranty provided that these products are suitable for any particular purpose or operational situation.
Polygon certifies that their product will be free from material defect. Polygon will not accept any liability for loss, damages, or costs from use or misuse of our products.
Specifications are subject to change, and may be affected by our continual process of improvement. Changes may be made without prior announcement.